Hewlett Packard 1741A Oscilloscope 100MHz *AS-IS / FOR PARTS* Bad Display
This unit appears to have a bad display. Light 'bleeds' from any illuminated sections and will persist on screen. The CRT may be going bad or the trigger function may be failing. The calibration voltage is working correctly and both Channels A & B correctly display the 1 Volt DC readings when dialing in the VOLT/DIV and TIME/DIV knobs. This unit is being listing AS-IS / FOR PARTS due to its partial functionality. Please view our photo set and contact us with any questions.